View Entire Collection
By Clinical Topic
By State Requirement
Diabetes – Summer 2012
Future of Nursing Initiative
Heart Failure - Fall 2011
Influenza - Winter 2011
Nursing Ethics - Fall 2011
Trauma - Fall 2010
Traumatic Brain Injury - Fall 2010
Fluids & Electrolytes
Learn how PbtO2 technology lets you accurately assess cerebral oxygenation at the bedside.
If you're caring for a patient with a severe brain injury, early detection of cerebral hypoxia and impending ischemia are key to preventing secondary brain injury. Neurologic monitoring at the bedside includes cerebral and systemic hemodynamic parameters. Continuous intracranial pressure (ICP) monitoring is the gold standard for assessing brain injury and the patient's response to treatment, but research has shown that secondary cerebral hypoxia and infarction can develop even when ICP and cerebral perfusion pressure (CPP) readings are normal.
Monitoring partial pressure of oxygen in brain tissue (PbtO2) can be used to collect more accurate and timely information about cerebral oxygen delivery and demand. In this article, we'll explain how using PbtO2 in conjunction with other assessment parameters can help you protect a patient from a secondary brain injury. (See Tracking trends in your patient's condition.)
Because the brain has very limited oxygen and glucose reserves, it depends on continuous and adequate blood flow to deliver oxygen and glucose to meet its metabolic demands. An acute brain injury can both disrupt normal oxygen delivery and increase metabolic demand, leading to secondary damage from the resulting cerebral edema and ischemia.
Because ICP and CPP values don't always accurately reflect tissue oxygenation in the injured brain, PbtO2 can provide crucial information. It's especially useful in managing the acute injury phase, when secondary brain injury is most likely to occur.
Patients who sustain a severe traumatic or neurologic brain injury (such as stroke or aneurysm rupture) and who are at risk for further injury from hypoxia and cerebral ischemia are candidates for PbtO2 monitoring. At our facility, patients who qualify for brain tissue oxygen monitoring have a Glasgow coma scale score of 8 or less (on a scale of 3 to 15, with 15 being the best), are mechanically ventilated, and require standard neurologic and systemic hemodynamic monitoring.
Before any invasive neurologic monitoring device is inserted, any coagulopathies must be corrected. Contraindications to brain tissue oxygen monitoring include coagulopathy and insertion site infection.
At our facility, we use the Licox CMP Triple Lumen Monitoring System (Integra NeuroSciences, Plainsboro, N.J.) for PbtO2 monitoring. The device measures PbtO2, brain tissue temperature, and ICP. It consists of a bedside monitor and display screen connected to two probes to measure brain tissue oxygenation and temperature. A separate ICP monitor is connected to an ICP probe.
The physician drills a single burr hole and places an intracranial bolt, then inserts each probe through the dedicated port of the Licox housing system. Probe placement is based on the patient's condition, a review of computed tomography (CT) scans, and therapy goals. If therapy is aimed at monitoring oxygen availability to damaged brain tissue, probes may be placed near (but not in) a cerebral lesion, avoiding areas of infarct or hematomas. If the probes can't be placed on the side of injury (for example, because of a postoperative bone flap), they're placed in the opposite hemisphere and used as a measure of global oxygenation.
Once the probes are in place, the physician applies a sterile dressing over the bolt site. The probe cables are connected to the monitor. The system is quickly and easily calibrated with a smart card.
After 10 to 120 minutes, when the brain tissue has stabilized from the microtrauma of probe insertion, the system continuously records and displays local cerebral oxygen and temperature measurements. The ICP probe is zeroed in the usual fashion and provides ICP data immediately. Some patients also may have a ventriculostomy for cerebrospinal fluid drainage.
After brain tissue has had time to stabilize following probe insertion, PbtO2 measurements can be recorded and tested. Perform an oxygen challenge test, particularly if the PbtO2 reading is unexpectedly low or you question the probe's accuracy. To perform the oxygen challenge test, place the ventilator's FIo2 setting on 100% for 2 to 5 minutes. We've found that 2 minutes of 100% FIo2 is adequate for testing probe sensitivity to oxygen, especially when FIo2 requirements are less than 80%. An accurate probe will reflect an increase in PbtO2. If the patient's PbtO2 doesn't respond to the increased FIo2, obtain a head CT scan to confirm correct probe placement. If the probe has been placed incorrectly, it must be removed and replaced at a new site through a new burr hole.
A major advantage of PbtO2 monitoring for patients at risk for cerebral ischemia is that it detects brain hypoxia as it occurs, so clinicians can intervene immediately to prevent secondary brain injury. Normal PbtO2 values range from 25 to 50 mm Hg; values below 15 mm Hg indicate ischemia, and values below 5 mm Hg indicate brain cell death.
The oxygenation value reflects oxygen delivery to the tissue around the probe and is interpreted as a marker of cerebral perfusion. Brain tissue temperature is also monitored because this value is needed to calculate the PbtO2. Also, systemic temperature doesn't always reflect brain temperature, which directly influences cerebral oxygenation and demand. A change of 1.8[degrees] F (1[degrees] C) in brain temperature can alter cerebral metabolism by 5% to 13%, affecting cerebral blood flow and ICP.
Brain tissue oxygenation values are just one piece of the clinical picture and shouldn't be used as the sole basis for diagnosis or treatment decisions. Interpret PbtO2 values in the context of other clinical findings. (See Responding to changes in PbtO2 values.)
The Licox system is compatible with CT scanning, but not with magnetic resonance imaging (MRI). The MRI technician can perform a wand scan to detect incompatibilities, but the entire bolt system may need to be removed.
Monitor the insertion site for signs and symptoms of infection and contusion. However, these complications are rare.
Monitoring for signs of secondary brain injury improves survival rates and helps minimize permanent neurologic deficits. Adding PbtO2 monitoring to traditional neurologic monitoring parameters provides more detail to the clinical picture and has proven safe and effective for patients with severe brain injury.
A key to preventing secondary brain injury is to limit the size of the initial injury. The following assessment values help you spot trends in your patient's condition.
* Partial pressure of oxygen in brain tissue (PbtO2)-normal range, 25 to 50 mm Hg. Values below 15 mm Hg indicate ischemia. Follow trends and intervene if values drop below 20 mm Hg.
* Intracranial pressure (ICP)-normal, less than 10 mm Hg. A value between 10 and 20 mm Hg is mildly to moderately elevated; above 20 mm Hg is severely elevated.
* Mean arterial pressure (MAP)-normal range, 70 to 110 mm Hg. A MAP below 60 mm Hg may indicate decreased cerebral perfusion pressure (CPP).
* CPP-normal range, 70 to 100 mm Hg. This value is calculated by subtracting ICP from MAP; a value below 50 mm Hg indicates compromised cerebral blood flow.
* End-tidal carbon dioxide (EtCO2)-normal range, 30 to 40 mm Hg
* Paco2-normal range, 35 to 45 mm Hg. A value less than 35 mm Hg may trigger cerebral vasoconstriction, reducing oxygen delivery to the brain. Keep Paco2 in the normal range until you can determine how it affects PbtO2.
* Pao2-normal value, 80 to 100 mm Hg. Decreased Pao2 can cause tissue hypoxia and reduced PbtO2.
* Jugular venous oxygen saturation (SjvO2)-normal range, 60% to 70%. Low values indicate a high extraction of oxygen characteristic of ischemia. However, problems with SjvO2 technology limit its usefulness as a sole measure of cerebral oxygenation. Also, this technology won't identify regional ischemia, so tissue hypoxia may be present despite normal or increased SjvO2 values; some causes of tissue hypoxia are: decreased blood flow (ischemic hypoxia), low Pao2 (hypoxic hypoxia), low hemoglobin concentration (anemic hypoxia), and hypermetabolic hypoxia caused by mitochondrial dysfunction and increased demand of cerebral tissues. Also, the tendency of catheters to migrate to the vessel wall and value errors, which require frequent blood sampling to calibrate the monitor, can limit the usefulness of SjvO2 values.
Here's how to respond to various causes of changing brain tissue oxygenation (PbtO2) values.
Because of increased oxygen demand due to:
* increased intracranial pressure. Treat with diuretics, cerebrospinal fluid drainage, sedation, craniotomy.
* pain. Administer analgesics.
* shivering. Administer meperidine, chlorpromazine, or a paralytic.
* agitation. Administer a sedative.
* seizures. Administer a benzodiazepine and adjunct anticonvulsive agent.
* fever. Administer acetaminophen or a nonsteroidal anti-inflammatory drug or use a temperature-regulating blanket.
Because of decreased oxygen delivery due to:
* hypotension. Administer isotonic fluids, such as 0.9% sodium chloride solution or hypertonic saline, and vasopressors.
* hypovolemia. Administer isotonic fluids, such as 0.9% sodium chloride solution or hypertonic saline, and blood replacement.
* anemia. Administer blood replacement.
* hypoxia. Increase FIo2 and positive end-expiratory pressure.
* sedatives, anesthesia, or paralysis. Reduce sedation, anesthesia, or paralysis as needed. Treatment may not be necessary.
* hyperdynamic state. Administer hyperventilation.
* hypothermia. Perform interventions to restore normothermia.
Eileen Maloney Wilensky is director of the clinical research division in the department of neurosurgery at the University of Pennsylvania Medical Center in Philadelphia, Pa. Stephanie Bloom is a research nurse in the department of neurosurgery at the University of Pennsylvania Medical Center.
Fitch W. Brain metabolism. In Cottrell J, et al. (eds), Anesthesia and Neurosurgery, 4th edition. St. Louis, Mo., Mosby, Inc., 2001.
Littlejohns L, Bader M. Guidelines for the management of severe head injury: Clinical application and changes in practice. Critical Care Nurse. 21(6):48-55, December 2001.
Sullivan J. Jugular venous oxygen saturation monitoring: Insertion (assist), care, troubleshooting, and removal. In Lynn-McHale D, Carlson K (eds), AACN Procedures Manual for Critical Care. Philadelphia, Pa., W.B. Saunders Co., 2001.
van den Brink W, et al. Brain oxygen tension in severe head injury. Neurosurgery. 46(4):868-878, April 2000.
Sign up for our free enewsletters to stay up-to-date in your area of practice - or take a look at an archive of prior issues
Join our CESaver program to earn up to 100 contact hours for only $34.95
Explore a world of online resources
Back to Top