View Entire Collection
By Clinical Topic
By State Requirement
Faith Community Nursing
Future of Nursing Initiative
A CRITICALLY ILL patient needed an intra-aortic balloon pump (IABP) to provide circulatory support. When the pump's helium tank was low, a nurse switched the pump off for 30 minutes to replace the tank.
At 2:30 a.m. the following day, the pump alarm sounded several times. The pump was shut off, and the patient was given heparin and blood products.
At 8:40 a.m., the physician removed the intra-aortic balloon (IAB) catheter without difficulty but found that no balloon membrane material was left on the catheter. The patient was immediately taken to the OR for surgery to remove the balloon membrane but later died.
Intra-aortic balloon pump counterpulsation is frequently used as therapy for patients with refractory left ventricular heart failure or for other indications. Benefits of IABP therapy include increased cardiac output, coronary blood flow, and systemic perfusion pressure and decreased afterload and myocardial oxygen demands.
Letting an IAB remain dormant for longer than 15 minutes is hazardous, not only because the patient loses valuable cardiac support, but also because blood becomes trapped within the folds of the deflated balloon, promoting the formation of clots. The IAB in this case was allowed to remain dormant twice.
As a nurse, you can help mitigate complications from IABP therapy. Suggested device management and precautions include the following:
[black right pointing small triangle] Frequently monitor the patient's physiologic responses to IABP therapy such as cardiac output, cardiac index, systemic vascular resistance, blood pressure, stroke volume, central venous pressure/right atrial pressure, and pulmonary capillary wedge pressure.
[black right pointing small triangle] Frequently check pump activity, helium level, driveline, and catheter.
[black right pointing small triangle] Replace empty helium tanks within 15 minutes-the sooner, the better. Keep extra helium tanks nearby to minimize the time the IAB is dormant.
[black right pointing small triangle] Ensure that additional fully operational pumps are readily available in case of an emergency. If an alarm sounds, check the patient first, then the pump. Considered a high-priority alarm, a helium leak alarm indicates a potentially serious condition warranting immediate attention. If you can't quickly identify the cause of the alarm, obtain another pump console. If the alarm continues to sound, notify the physician immediately. The balloon may have ruptured, requiring immediate removal.
[black right pointing small triangle] If pump operation can't be restored within 15 minutes, the balloon becomes dormant. You must manually inflate and deflate the balloon optimally every 5 minutes; at a minimum, at least several times per hour. To reduce the risk of thrombus formation, continue this until a new pump is in place or the balloon is removed. Follow the IAB product instructions to check for blood in the driveline before starting manual inflation and deflation.
[black right pointing small triangle] Blood in the driveline or catheter can indicate that the IAB has ruptured. Notify the physician, who must remove a ruptured IAB immediately.
[black right pointing small triangle] Never inject air into the central (female luer fitting) lumen; you could introduce an air embolus into the patient's circulation.
[black right pointing small triangle] If your patient has problems with either the IAB or the pump, quarantine the device after it's removed and notify your biomedical engineering department so that the device can be returned to the manufacturer for examination and evaluation.
[black right pointing small triangle] Always follow the standards of clinical practice and your hospital's policies and procedures when you care for a patient receiving IABP therapy.
For life-long learning and continuing professional development, come to Lippincott's NursingCenter.
Debunking Three Rape Myths
Journal of Forensic Nursing, October/December 2014
Expires: 12/31/2016 CE:2.5 $24.95
Drug updates and approvals: 2014 in review
The Nurse Practitioner, 13December 2014
Expires: 12/31/2016 CE:3 $27.95
Can Food Processing Enhance Cancer Protection?
Nutrition Today, September/October 2014
Expires: 10/31/2016 CE:2 $21.95
More CE Articles
Subscribe to Recommended CE
Differential Diagnosis of High Peak Airway Pressures
Dimensions of Critical Care Nursing, January/February 2015
Free access will expire on February 2, 2015.
The Institute of Medicine’s 2014 Committee on Approaching Death Report: Recommendations and Implications for Nursing
Journal of Hospice and Palliative Nursing, December 2014
Free access will expire on January 19, 2015.
A missed connection: Depression screening in cardiac inpatients
Nursing2014 , December 2014
Free access will expire on January 19, 2015.
More Recommended Articles
Subscribe to Recommended Articles
Back to Top