View Entire Collection
By Clinical Topic
By State Requirement
Diabetes – Summer 2012
Future of Nursing Initiative
Heart Failure - Fall 2011
Influenza - Winter 2011
Nursing Ethics - Fall 2011
Trauma - Fall 2010
Traumatic Brain Injury - Fall 2010
Fluids & Electrolytes
Certain needleless intravascular (I.V.) catheter access components significantly raise the risk of bloodstream infections, according to a recent report. Clinicians at the University of Nebraska Medical Center in Omaha noticed a dramatic rise in bloodstream infections following adoption of SmartSite Plus, a positive-displacement, luer-activated needleless I.V. connector valve. This type of connector expels a small amount of flush solution into the catheter when the device is disconnected to clear the catheter and prevent retrograde flow.
In the intensive care unit, the infection rate nearly tripled after this device was introduced. Similarly, the bloodstream infection rate in regular nursing units more than doubled. After the hospital stopped using these devices, infection rates returned to prior levels.
Researchers speculate that microbes and debris easily accumulate on the device's shallow depression and rim, inviting infection. The device's movable parts also introduce irregularities in fluid flow that may promote stagnation, creating favorable conditions for microbe growth. And the device is opaque, preventing visual inspection for problems.
Every hospital in the country is using similar devices, the researchers say. They've notified the Food and Drug Administration and the device's manufacturer (Alaris Products, CardinalHealth) of their findings.
Rupp ME, et al., Outbreak of bloodstream infection temporally associated with the use of an intravascular needleless valve, Clinical Infectious Diseases, June 1, 2007.
Sign up for our free enewsletters to stay up-to-date in your area of practice - or take a look at an archive of prior issues
Join our CESaver program to earn up to 100 contact hours for only $34.95
Explore a world of online resources
Back to Top