View Entire Collection
By Clinical Topic
By State Requirement
Diabetes – Summer 2012
Future of Nursing Initiative
Heart Failure - Fall 2011
Influenza - Winter 2011
Nursing Ethics - Fall 2011
Trauma - Fall 2010
Traumatic Brain Injury - Fall 2010
Fluids & Electrolytes
A 58-year-old man undergoing surgery experienced a sudden decrease in arterial pressure for no apparent reason. After he was treated with vasopressors and his blood pressure stabilized, the staff realized that a nitroglycerin infusion had flowed into his I.V. access via gravity and the bottle was empty. Noticing a gap at the top of the infusion pump door, they found that the upper flange part of the tubing had been loaded improperly, disarming the mechanism that prevents free-flow. The pump database revealed that the pump had alarmed and displayed an error message during setup, but the message reported a possible occlusion, so the free-flow problem wasn't recognized and the alarm was overridden.
In a simulated clinical setting, investigate how improperly loaded tubing could affect the performance of your infusion pumps. If you uncover problems, report them at email@example.com. All reports will be forwarded to the Food and Drug Administration and the device manufacturer. Your facility may also want to contact its infusion pump vendor to see if representatives are aware of such risks with their pumps.
Schroeder ME, et al. Tubing misload allows free flow event with smart intravenous infusion pump. Anesthesiology. 105(2):434-435, August 2006.
Sign up for our free enewsletters to stay up-to-date in your area of practice - or take a look at an archive of prior issues
Join our CESaver program to earn up to 100 contact hours for only $34.95
Explore a world of online resources
Back to Top