View Entire Collection
By Clinical Topic
Diabetes – Summer 2012
Future of Nursing Initiative
Heart Failure - Fall 2011
Influenza - Winter 2011
Nursing Ethics - Fall 2011
Trauma - Fall 2010
Traumatic Brain Injury - Fall 2010
Fluids & Electrolytes
For the purpose of this document, SGNA has adopted the following definitions:
Critical Medical Devices refer to those instruments that may be introduced directly into the bloodstream or into other normally sterile areas of the body. These devices break the mucus membrane (American Society for Testing and Materials [ASTM], 2000) and/or come into contact with sterile tissue or the vascular system.
Endoscopic Accessory Instruments refer to medical instruments designed for insertion into a flexible endoscope (ASTM, 2000). These are devices used during endoscopy other than the endoscope. These include, but are not limited to, biopsy forceps, snares, bite blocks, guide wires, irrigation tubes, and dilators. These devices may or may not have lumens, porous or loosely joined surfaces, or access ports for flushing; and may or may not be capable of being completely disassembled during reprocessing.
Reprocessing refers to the sequence of cleaning, lubricating (if necessary), and sterilizing (ASTM, 2000) or high-level disinfecting steps that will assure an endoscopic accessory is patient-ready (SGNA, 2003).
Reusable Device refers to an instrument designed and validated by the manufacturer to be used more than once, provided that after each use, an appropriate reprocessing protocol and functionality check is performed following manufacturers' recommendations (ASTM, 2000; SGNA, 2003).
Single Use Device (SUD) refers to an instrument designed for one-time use only, on one patient, during a single procedure. SUDs are not designed by their manufacturers to be reprocessed and/or used on another patient. The manufacturer's labels on these devices do not include reprocessing instructions, and may or may not identify the device as single use (United States Food and Drug Administration [U.S. FDA], 2000). SUDs are also referred to as disposable devices (U.S. FDA, 2000).
Valves refer to the air/water valve, suction valve, and biopsy port cover to the flexible endoscope (ASTM, 2001).
Proper reprocessing of endoscopic accessories and valves is critical to the safe and successful treatment of patients (Alvarado, Reichelderfer, and the Association for Professionals in Infection Control and Epidemiology [APIC] Guidelines Committees, 2000; SGNA, 2005). SGNA supports increased research in the areas of accessory and endoscope design in an effort to manufacture devices that can be easily disassembled, cleaned, high-level disinfected and/or sterilized.
The United States Food and Drug Administration requires the manufacturers of reusable devices to provide instructions for cleaning and high-level disinfection or sterilization (U.S. FDA, 1996).
SGNA supports the following positions:
A. All accessories labeled as reusable must be reprocessed according to manufacturer's instructions.
B. Accessories that are classified as critical medical devices require sterilization.
C. Critical items labeled for single-use should not be reprocessed and/or reused (SGNA, 2002).
D. Following each use of the endoscope, valves must be removed, manually cleaned, and high-level disinfected or sterilized according to the original equipment manufacturer's instructions. This must occur as part of the cleaning and disinfecting process for the endoscope.
E. When using an automated endoscope reprocessor (AER), the AER manufacturer's instructions regarding reprocessing of valves must be followed.
Alvarado, C. J., Reichelderfer, M., & The Association for Professionals in Infection Control and Epidemiology Guidelines Committees. (2000). APIC Guideline for infection prevention and control in flexible endoscopy. American Journal of Infection Control, 28, 138-155. [Context Link]
American Society for Testing and Materials. (2000). Standard practice for reprocessing of reusable, heat-stable endoscopic accessory instruments (EAI) used with flexible endoscopes [Standard]. West Conshohocken, PA: Author. [Context Link]
American Society for Testing and Materials. (2001). Standard practice for cleaning and disinfection of flexible fiberoptic and video endoscopes used in the examination of the hollow viscera [Standard]. West Conshohocken, PA: Author. [Context Link]
Society of Gastroenterology Nurses and Associates, Inc. (2002). Reuse of single-use critical medical devices [Position Statement]. Chicago: Author. [Context Link]
Society of Gastroenterology Nurses and Associates, Inc. (2003). Gastroenterology nursing: A core curriculum (3rd ed.). Chicago: Author. [Context Link]
Society of Gastroenterology Nurses and Associates, Inc. (2005). Standards of infection control in reprocessing of flexible gastrointestinal endoscopes [Standard]. Chicago: Author. [Context Link]
United States Food and Drug Administration. (1996). Labeling reusable medical devices for e facilities. Retrieved September 30, 2005, from http://fda.gov/cdrh/ode/198.pdf. [Context Link]
United States Food and Drug Administration, U.S. Department of Health and Human Services, Center for Devices and Radiological Health. (2000). Guidance for industry and for FDA staff: Enforcement priorities for single-use devices reprocessed by third parties and hospitals [Guidance]. Rockville, MD: Author. Retrieved September 30, 2005, from http://www.fda.gov/cdrh/reuse/1168.html. [Context Link]
Copyright (C) 2006, Society of Gastroenterology Nurses and Associates, Inc. (SGNA).
This document was prepared and written by the SGNA Practice Committee and adopted by the SGNA Board of Directors in May 2002. Revised in October 2005.
Additional copies can be printed from SGNA's website at http://www.sgna.org.
Susan Bohlander, BSN, RN, CGRN, Chairperson
Cynthia M. Friis, MEd, RN-BC
Donna Girard, BSN, RN, CGRN
Anne Scroggs, MSN, APRN-BC
LeaRae Herron-Rice, BSN, RN, CGRN
Loralee Kelsey, RN, CGRN
Carol Kraai, MSN, RN, CGRN
Lisa D. Miller, LPN, CGN
Cindy Taylor, MSA, BSN, RN, CGRN
Trina Van Guilder, BSN, RN, CGRN
Sign up for our free enewsletters to stay up-to-date in your area of practice - or take a look at an archive of prior issues
Join our CESaver program to earn up to 100 contact hours for only $34.95
Explore a world of online resources
Back to Top